David Pilling 
Main »
Phase Noise aka Jitterstarted 26th July 2016  completed 23rd September 2016 Quoting Wikipedia "phase noise is the frequency domain representation of rapid, shortterm, random fluctuations in the phase of a waveform, caused by time domain instabilities (jitter). Generally speaking, radio frequency engineers speak of the phase noise of an oscillator, whereas digital system engineers work with the jitter of a clock." I measured the time between output pulses using an Arduino Pro Mini based on the ATMega 328p, this has a timer capture register, so results should not be affected by interrupts. The clock is a crystal oscillator running at 16 MHz, which (see GPS) is likely to be accurate to better than a cycle per second. The PUT has an output which goes negative in around 1 μs, to match it to the ATMega 328p an inverting single transistor buffer is needed. The ATMega is set up to capture the time of the resulting positive edge. The point being that the pulses have fast rise times and there's not another lot of uncertainty around external comparators and their power supply voltages. The variations in period are visible on a 'scope. Graphics show PUT oscillator (I used a 5V supply and a 10nF capacitor, the output is taken from the junction of R1 and R2), histograms of output inverse time and time and a scatter plot of inverse time of cycle N against inverse time of cycle N+1. The histograms do not look very random. As a crude test I plotted the inverse time of cycle N against that of cycle N+1. It seems that a short cycle is followed by a long one. I found a similar histogram in [4], which says it represents "deterministic jitter". One possibility I could imagine is that a long cycle pulls down the power supply voltage a little and results in a following short cycle. I found no sign that this was going on, but the general idea that after a long cycle the circuit is in a state where the next cycle will be short seems plausible. In other words the circuit has to have a memory of the previous cycle. After a while I noticed that the frequency of the oscillator happens to be nearly 100 Hz  the mains here is 50 Hz  allowing alternate half cycles to push and pull the oscillator frequency. I built the circuit on a typical solderless breadboard; the connection between the collector of Q1 and the base of Q2 was three inches of insulated wire; replacing it with a much shorter connection made the double peaks less prominent whilst holding the wire between my fingers made the jitter worse. This point in the circuit has high impedance when the transistors are turned off. Where does the characteristic shape of the histogram come from? Imagine that the oscillator is going a lot faster than the 50 Hz interference. Effectively an oscillation takes place with a fixed value of the interference. If that oscillation coincides with the maximum value of interference then the oscillation will have a minimum time, the other extreme of interference will give a maximum time. Given a sine wave (like the 50 Hz mains) for most of the time values are near one extreme or the other. The histogram is showing the probability distribution for a sine wave. The links [5] [6] [7] [8] confirm this idea. Another way of putting this is that the oscillator is sampling the sine wave with the period of the oscillator being the value. Plotting the period of oscillation against time allows the interfering signal to be reconstructed  see later. Looking at the PUT circuit above, if the voltage across C1 follows the usual RC equation (charging from a voltage Vs): ` V=V_s(1exp(t/(R_3C_1))) `
and the threshold at which the transistors turn on at is Vt, then the period will be: ` T=R_3C_1 ln(1V_t/V_s) `
For Vt/Vs=0.5 the ln is 0.7, for Vt/Vs=0.99 the ln is 4.6 and for Vt/Vs=0.999 the ln is 6.9 meaning that the period will not increase much even when Vt and Vs are within millivolts of one another. There's a bit of messing about with Vbe  roughly the lowest voltage across C1 and the threshold is Vbe above the voltage at the junction of R1 and R2. As Vt approaches Vs the period gets longer. The rate of change of voltage with respect to time is: ` exp(t/(R_3C_1))/(R_3C_1) `
For large times the voltage does not change much. The rate of change of period with respect to changes in the threshold voltage is: ` ((R_3C_1)/V_s)//(1V_t/V_s) `
When ` V_t ` approaches ` V_s ` period becomes sensitive to changes in ` V_t `. All of which mean that a threshold far away from the supply voltage will be cut by a rapidly changing voltage and timings are relatively insensitive to changes in the threshold due to noise. Whilst a threshold close to Vs has the opposite behaviour. For an oscillator with little jitter pick a threshold away from Vs, and for more jitter chose Vt close to Vs. One can vary Vs as easily as Vt. Three sets of graphics for Vs=5, 4 and 3V. Each set is histogram of time, inverse time, inverse timr N against N+1, inverse time N against N+1, reconstructed 50 Hz signal, and (jitter) frequency spectrum. For the reconstruction, the time duration of oscillation is being plotted against run time modulo 20 ms. For the frequency spectrum the time between pulses is used as a sample value and the time the pulse occurs at as the sample time. The last two are an approximation, reasonable if the oscillation period jitter is small compared to the period of oscillation. This time I used a 1nF capacitor. Download 5V data Download 4V data Download 3V data It is apparent there is 50 Hz interference. Again the circuit is on a solderless bread board, but with no long wires. It's just very sensitive. As Vs decreases the time of oscillations increases whilst the spread in time increases showing the greater sensitivity to noise. Below 3V it gets difficult to adjust the voltage by hand, a little too low and oscillation stops.
The 50 Hz interference is apparent; see later for interpreting Allan deviation plots. The graphs were plotted using "allantools"[13]. [14] takes the point of view of how error varies with changing numbers of measurements. As the number of measurements increases the effects of noise cancel out. The Allan plot allows finding the number of measurements that minimises the error. In the Allan plot for 5V above the first dip is at around 10 samples, (at 529 Hz) 20 ms worth of oscillations, one mains cycle. There are many ways of producing periodogram plots or finding weak cyclical signals in noise [18] [19]. Self adjusting PUT oscillator The obvious implementation is to feed the output of the oscillator into a frequency to voltage converter and use the resulting voltage for Vs. However there is a simple addition to the PUT which gives the same effect which is shown below. Graphics are the self adjusting PUT circuit, histogram of time, inverse time, inverse time of cycle N against cycle N+1. LT Spice file of the circuit Compared to the standard PUT oscillator C1 will charge up to close to Vt almost as quickly but it will then spend a long time just below Vt. Graphics show the voltage across C1 for the self adjusting PUT oscillator and the standard version. For the former the simulation gives a gradient of around 6 mV per second  if it was true and if one could measure time to 1 μs then sensitivity would be in picoVolts, down amongst thermal noise levels. Graphics show the LT Spice results for the self adjusting PUT oscillator using the same values as above. On the bright side, the 50 Hz interference has gone from these results (it's easy to simulate if desired) but they raise the problem of where all the noise is coming from  insignificant changes to the LT Spice model change the results, apparently then numerical noise.
This did stop the interference, the results for the simple PUT oscillator effectively collapsed to a single frequency; variation in time was only as big as the 1/16 th μs resolution of the microcontroller. Graphs show time between pulses, inverse time of N against N+1 and Allan deviation. In contrast the self adjusting PUT oscillator produced a nice spread of times between pulses. Graphs show time between pulses, inverse time of cycle N against N+1 and Allan deviation (overlapping, overlapping with gradients and modified with gradients).
From the graphics it appears that the self adjusting PUT oscillator sets off with white phase noise and ends with frequency drift. In the middle there is a range of random walk phase variation or white frequency noise. Phase is the time integral of frequency. A random walk is the summation of independent, identically distributed random variables. It is possible to calculate the Allan deviation from the Power Spectral Density (PSD), but not the other way around, gradients only show what a given PSD would produce an Allan deviation looking like. I worried if my use of Allan variance was correct. The discussions of Allan variance are presented from the point of view of someone with a 10 MHz oscillator and a frequency counter with a 1 second gate time. Every second they get a value for the frequency. They can compare one frequency value with the next. The point being the time between values is constant at one second, and the duration of comparisons can be varied in multiples of it. Here the time between samples is not constant because it is the variable duration of the cycles. I am measuring the duration of single cycles not millions. The waveform is not sinusoidal and I can't measure the phase within a cycle, all I have are the times of events. I came to the conclusion the results are valid. With Allantools there is a program which plots various types of noise ("noise color demo") allowing the Allan deviation results for each to be seen. For white noise it generates a series of random numbers and feeds them into the code for calculating the Allan deviation. There's no need for the complexity of equal sample times etc. I also tried sampling my data with equal times (counting the number of cycles in every so long) and got the same result (phase random walk) although I lost the detail for times less than the sample time. [29] says that the scatter plot of the form shown in the second graphic of the previous set is indicative of white phase noise.
As ever the 5V data have periodic interference and fail the Ljung Box test. The self adjusting PUT also fails. It is apparent there is a big correlation between neighbouring cycles (lag==1), for example a short period being followed by a long one. It puzzled me why the correlation is always around 0.5. Eventually I read in [28] ("5.5.4. The Lag 1 Autocorrelation") that this value is indicative of white phase noise  agreeing with the Allan deviation. (see Time Series for a derivation of the value). Extracting every Nth data value (with N above 10) gives a series which does pass the test. So as the Allan deviation suggests, at long enough times there is randomness in frequency. Calculating the 'phase', i.e. going back to the times of occurrence of pulses as opposed to the difference between them, also results in data which passes the test.
A conclusion is that if a circuit has sensitivity to noise it is likely to also have sensitivity to external noise which may not be random. Is there frequency locking? Yes where the period is greater than that of the interference, the capacitor voltage is just below the threshold for a long time, it is likely that a peak of the interference will push it over the threshold, so the period locks to a multiple of that of the interference. In the other situation of oscillation period less than that of the interference, locking did not seem to occur. Pot twiddling it was difficult to make the oscillator work at exactly 50 Hz. For a lock to occur not only does the frequency have to be the same, but the phase has to be correct too, because period depends on interference phase. Half the interfering sine wave will work to increase errors in frequency (the other half decreases them). Perhaps this is a "phase antilock loop". Software Download
References

David Pilling's Wiki Set view 
Page last modified on June 12, 2018, at 12:59 AM  Powered by PmWiki (pmwiki2.2.131) 